Short video to explain how and when to replace soda lime on Moduflex™ veterinary anesthesia machines. Video transcription Hi, welcome to another Dispomed video. Today we will be looking at how and when to replace the sodasoarb in your anesthesia machine. Soda lime is used to absorb the CO2. There is a colour indicator included … [Continue Reading]
Extend the life of your handpieces! Watch this video to learn when and how to lubricate the handpieces of your dental unit. When properly lubricated, they are easier to use and have a longer life expectancy. Video Transcription Welcome to a new series of videos by Dispomed, this series will cover dental equipment. Today, we … [Continue Reading]
Video transcription Hi everyone, welcome to another Dispomed video. Today, we are going to talk about the differences between the standard Y circuits and the Coaxial circuits. Everyone is familiar with the standard Y circuits. The Y circuit has two independent connections: one for inspiration and another for expiration. Both of them are 22-mm connectors. … [Continue Reading]
Are your tips not working as well as before? The time may have arrived to replace them. Watch this short video which explains how and when to change your tips and burs on your handpieces.
Make sure that your Highdent™ dental unit is in perfect condition for all the dental cases planned in the next month! The dental month is just around the corner and you will certainly use your Highdent™ dental unit more frequently. Take the time to watch this video to remember how to perform a maintenance on … [Continue Reading]
Learn about a simple and easy method to verify your veterinary anesthesia machine before each procedure. Our technician will show you, step by step, how to make sure your machine is in working order. Following these easy steps will reduce the risk for your staff and your patients as well as helping you to foresee … [Continue Reading]
Do you have questions about how to fill or empty your key fill vaporizer? Watch this video and get useful tips from our expert on how to fill your TEC-3, TEC-4, and TEC-5 vaporizer.
Video Transcription Today we will talk about Patient Circuit Pressure Alarms, how to install and use them. Patient Circuit Pressure Alarms are used to help notify you in case of pressure reaching a dangerous point. They are easy to install on any type of anesthesia machine. All you have to do is hook up the … [Continue Reading]
Learn why the Moduflex™ Bain circuit adapter is the best open circuit for your surgeries on small animals. Video Transcription Today we’ll talk about the Moduflex Bain circuit adapter, its advantages, its operations and its use. We will also talk about the fact that it can be mounted on any type of anesthesia machine and … [Continue Reading]